To comply with the regulations for electric safety in medical environments (International Standards IEC) an additional equipotential bonding connector is provided.
In order to eliminate any difference in potential between touchable parts, ground and other conductive parts and to protect people from electric shock make sure that the support arm is connected to the potential bonding. Furthermore, consider the national electric safety regulations.
Mounting the Monitor
1. Ensure that the locking lever is engaged with the arm in the highest vertical position. Do not disengage the locking lever unless an instrument is mounted on the end of the support arm.
2. Slide the support arm into the ErgonoFlex wall channel or mount it to the European rail or pole and tighten the set screws.
3. Only after set screws have been tightened can medical devices be mounted. For most devices a specific adapter plate provided by ErgonoFlex med is needed.
Please refer to assembly instructions provided with the adapter plate.
4. Once the adapter plate has been attached to the device, secure it to the mounting plate of the support arm. Slide the device into the mounting plate until plunger locates in the clearance hole. Tighten the nylon screws in the mounting plate.
With VESA adaptations the flat panel is attached directly to the VESA mounting plate at the front of the support arm. Please use the M 4 screws with the suitable length. If needed nylon spacers can be used.
Tilt, Swivel and Pivot Adjustments
To adjust the overall tilt tension tighten or loosen the (2) hex screws with the hex key 4 mm provided.
To adjust the swivel and/ or pivot tension remove the end caps and tighten or loosen the (4) head screws.